MiniSpin® is a registered trademark of Eppendorf AG, Germany. .. >Read this operating manual completely before using the device for the first time. Also. Get Eppendorf MiniSpin plus Manual. Get all Eppendorf manuals!. Get Eppendorf MiniSpin plus Owner manual. Get all Eppendorf manuals!.
|Published (Last):||5 August 2006|
|PDF File Size:||1.31 Mb|
|ePub File Size:||2.76 Mb|
|Price:||Free* [*Free Regsitration Required]|
Centrifuge shakes during acceleration. Attendere che il rotore si fermi. Rotor care instructions Rotor use instructions Inspection Seal replacement Lubrication Function test. Fasten the rotor lid and close the centrifuge lid. No main power connection. Rotor not loaded symmetrically. Rotor Assurance Plan – Visual rotor checkacc. Stop centrifuge run and load centrifuge symmetrically.
Eppendorf MiniSpin plus User Manual
Instrument set-up Initial operation Instruction of user Checking for damage Checking lid latch Checking speed sensor Verification of operating parameters acc. Or point us to the URL where the manual is located. Our website does not support your outdated browser version.
Forgot password or username? Disconnect the centrifuge from the mains supply, remove the rotor and clean it separately.
The nominal values of the last run appear in the minispih. MiniSpin plus MiniSpin 14, rpm 13, rpm Max. Rotor not loaded symmetrically. Sbloccaggio di emergenza v. Disconnect the centrifuge from the mains supply, remove the rotor and clean it separately.
Inspect the rotor lid regularly. If the centrifuge makes unusual noises when started, the rotor or rotor lid is not fastened correctly. MiniSpin plus MiniSpin 14, rcf 12, rcf Max. The rotor, inner centrifuge lid, and lid latch are made of metal for maximum operation safety.
Please update to a current version, e. Extran neutral, RBS neutral. Eppendorf International Choose mmanual. Solo allora alzare la MiniSpin ed attraverso l’apertura nella piastra del fondo, spostare eppenodrf disco nella direzione della freccia, servendosi di una penna a sfera. Two models are available to meet your performance requirements. Brochure — Endless Possibilities. Fasten the rotor lid and close the centrifuge lid.
White Paper — Routine Maintenance of Centrifuges. Switch the device off immediately by pressing “STOP”. Your manual failed to upload Don’t see a manual you are looking for?
No main power connection.
Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge.
Centrifuge shakes during acceleration. Controllare il fusibile del laboratorio. Switch the device off immediately by pressing “STOP”. Real—life personal centrifuges do exist in the Minispin and MiniSpin plus from Eppendorf. Only eppensorf agents may be used for cleaning purposes e.
Eppendorf MiniSpin plus : Manual
To open the lid. With MiniSpin plus, the time can be pre-selected between 15 seconds and 99 minutes, and with MiniSpin between 15 seconds and 30 minutes. MiniSpin plus MiniSpin 14, rcf 12, rcf Max. Open tube lids can be torn off during centrifugation and damage the rotor lid or centrifuge.
MiniSpin plus provides speed sufficient for molecular biology separations. Instructions for use — Rotor F Wait for the rotor to come to a standstill. Be sure to close the tube lids tightly prior to centrifugation. Entry-level centrifuges that are powerful and easy-to-use. System test run OQ Report and certificate. Rotor is still spinning.
Tell us about it.
Eppendorf MiniSpin plus User Manual | 20 pages | Also for: MiniSpin
If this takes place during the run, the display switches back after 20 seconds. Extran neutral, RBS neutral. To open the lid.
Tell us what’s missing.