II. Improved generator4. Enlarged mm resection loop1. Safe bipolar technology Need a big reason to switch to the new. GYNECARE VERSAPOINT ™ II. Central Sterile · Surgical Instruments · Endoscopic Devices · Hysterscopes. Item Shown. GYNECARE VERSAPOINT Bipolar Electrosurgery System by Ethicon. MAUDE Adverse Event Report: ETHICON GYNECARE VERSAPOINT During hysteroscopy, the versapoint resectoscope was in the patient but not in use.
|Published (Last):||15 March 2018|
|PDF File Size:||18.78 Mb|
|ePub File Size:||9.88 Mb|
|Price:||Free* [*Free Regsitration Required]|
Attach the assembled working element, hysteroscope, and electrode assembly to the sheaths. The driver must depress the brake pedal in order to verapoint the gear selector from Park More information. Weak vacuum at wand. A Guide to Hysteroscopy. Item Qty Description Service More information.
Attach the light cable from the light source vrrsapoint the hysteroscope light post. Remove the U-bolt nuts More information. Although the install looks daunting at first, once you get the learning curve. You don t have to live with.
A NOTE indicates important information that helps. Attach the outflow tubing to the outflow value on the outer sheath. When tipping the Cable Processor Module back, after removing the toggle arm pinuse extreme caution not to drop More information.
GYNECARE VERSAPOINT Bipolar Electrosurgery System – PDF
Clogged hoses or wand tube. If your generator version supports both footswitch and possible future handswitchable accessories, fersapoint that the correct activation source is selected.
To disassemble the electrode elements, simply pull them apart. Purpose The purpose of this document is to describe the procedure necessary to replace one of the ball nuts.
Naturally, you may have questions about More information. Check insulation of connector cable.
GYNECARE VERSAPOINT Bipolar Electrosurgery System
Accessing the printer system board Accessing the printer system board 1 Turn the printer off. It is disclosed to the receiving More information. To ensure that your installation proceeds smoothly–please. Ensure ambient temperature is within operating vsrsapoint. Lever mounting base C. Level Detector Replacement More information. Dilate the cervix using an appropriate cervical dilator if necessary Insert the assembled standard obturator and sheaths into the uterus through the cervix Remove the standard obturator from the sheath by depressing the inner sheath-locking ring release button and withdrawing the obturator Inner Sheath Gynecaee Sheath Obturator Standard obturator not pictured Grey float Full Flush F.
Ethicon – Versapoint Community, Manuals and Specifications | MedWrench
To make this website work, we log user data and share it with processors. Gauge reads normal 10hg to 14hg 1. Do not insert sharp or tiny More information. If you gynecars questions when installing, operating or.
High Limit Switch Replacement E. Start display at page:. Use painter s tape rather than duct tape to secure window. It is a document to be used in conjunction. Never attempt to service. Remove connector and inspect pins for damage.
Plastic Replacement Sleeve 2. After replacing brake pads, depress brake pedal firmly More information. Case study Specifying durability and repair for laptop computers June A case study of a Hewlett-Packard notebook to illustrate and encourage the durability and repair of laptop computers.
Forums Documents and Manuals.
Avoid placing objects in the path of the blades. It is a document to be used in conjunction More information. Once display reads Connect Cable, connect hand piece to generator. The electrician More information.
Ethicon Gynecare 00482 VersaPoint Bipolar Electrosurgery Generator System
Replacing a Heating Element C. You can adjust your Community Subscriptions in Settings. Flush handle assembly D. Insert the standard obturator into the sheath Lock the standard obturator in place by pressing it firmly into the sheathlocking ring until the obturator snaps into place 4.